Research and Clinical Instruments Manufacturer & Supplier
Acquisition InterfacesNational Instruments USB-6341-BNCHEKA LIH 8+8 Data Acquisition SystemAmplifiersNeuroLog SystemD360 8-Ch. EP/EEG/EMGD360R 4 Channel Isolated Research Amplifier/FilterD440 2/4 Ch. EMG/EPHEKA EPC10 Patch ClampHEKA EPC10 2/3/4 Ch. Patch ClampHEKA EPC800 Patch ClampAnti-vibration SystemsThorLabs Science DesksNarishige ITS Anti-vibrationNarishige Double MagnetNarishige SBP-2 BaseplateCell InjectorsPLI-100A Pico-injectorPLI-10 Pico-injectorNarishige IM-11-2Narishige IMS-20Narishige IM-21Narishige IM-400Injection Accessories
Incubators & ChambersAutomate Perfusion ChambersMedical Systems MicroincubatorsSSD Brain Slice KeepersSSD Brain Slice ChambersIontophoretic DevicesD380 Dye MarkerMains Noise EliminatorsHumBug Noise EliminatorD400 Mains Noise EliminatorManipulatorsElectrophysiology System ManipulatorsInjection System ManipulatorsStereotaxic Manipulators“YOU” Compact ManipulatorsChronic ManipulatorsAccessoriesMicroscope Adaptors (Ephys)Microscope Adaptors (Injection)
MEA SystemsMED64 – BasicMED64 – Quad IIMED64 – AllegroMED64 – Plex 4/8MED64 – PrestoMED64 – Mobius SoftwareMED64 ProbesPerfusion EquipmentPerfusion SystemsPerfusion AccessoriesPipette FabricationMicropipette PullersMicroforges & MicrogrindersReplacement PartsCapillary GlassProgrammers & TimersDG2A Train Delay GeneratorNeuroLog System
Signal GeneratorsTG315 Function GeneratorSoftwareQtracW Threshold TrackingAutoMate EasycodeHEKA Chartmaster SoftwareHEKA Patchmaster SoftwareHEKA Fitmaster SoftwareStimulatorsDS2A Constant VoltageDS3 Constant CurrentDS4 Biphasic Constant CurrentDS5 Bipolar Constant CurrentDS7A/DS7AH Constant CurrentDS7R Constant Current ResearchDS8R Biphasic ResearchD121-11 Mounting FrameD185 Transcranial MultiPulseD330 MultiStim SystemNL800A Current Stimulus Isolator
A/D Interface ModulesNL201 – Spike TriggerNL601 – Pulse IntegratorAmplifier ModulesNL100AK – HeadstageNL100RK (NL100AKS & NL100C)NL102G – DC PreamplifierNL104A – AC PreamplifierNL106 – AC/DC AmplifierNL108A – Pressure AmplifierNL109 – Bridge AmplifierNL120S – Audio AmplifierNL820A – 4-Ch. IsolatorNL844 – 4-Ch. AC PreamplifierAnalogue ModulesNL254 – RatemeterNL506 – Analogue SwitchNL703 – EMG Integrator
Digital ModulesNL405 – Width/DelayNL501 – Logic GateNL505 – Flip FlopNL603 – CounterNL730 – Pulse ShiftFilter & Conditioner ModulesNL125/6 – Band-Pass FilterNL134/5/6 – 4-Ch. Low Pass FiltersNL143 – 3-Ch. Difference AmplifierNL144 – 4-Ch. High Pass FilterNL530 – Signal ConditionerNL540 – Inverting Attenuator (Alt. Gain)Generator ModulesNL301 – Pulse GeneratorNL304 – Period GeneratorNL412 – Pulse
NeuroLog AccessoriesAccessory KitsAdaptors & Adaptor CablesSockets (for cable mounting)Sockets (for panel mounting)Plugs (for cable mounting)Extension CablesCablesElectrode HoldersMiscellaneous AccessoriesNeuroLog System CasesNL900D – NeuroLog System CaseNL905 – Compact NeuroLog System Case
Pressure Transducers & AccelerometersPressure TransducersForce TransducersAccelerometersStimulator ModulesNL510 – Pulse BufferNL512 – Biphasic BufferNL800A Constant Current Stimulus Isolator
Application NotesSignal AmplificationTriggering & Pulse GenerationSignal Conditioning Filtering & ProcessingElectrical Stimulation
Isolated Amplifiers for EMG/EEG/EP D440 2/4-Ch. EMG AmplifierD360 8-Ch. Patient Amplifier D360R 4-Ch. Research AmplifierAmplifier Accessories D175 Electrode Impedance Meter D179 Performance Checker D360 Audio Interface D360 USB to Serial Adaptor D177 Bio-Feedback Unit
Peripheral Stimulators DS5 Isolated Bipolar Constant Current Stimulator DS7A & DS7AH HV Constant Current Stimulator DS7R HV Constant Current Research Stimulator DS8R Biphasic Constant Current Stimulator
Transcranial Cortical StimulatorsD185 MultiPulse Cortical Electrical StimulatorStimulator AccessoriesD188 Remote Electrode SelectorElectrode Connection HeadboxesTrigger CablesElectrode HandlesMiscellaneous Items
Neurodiagnostic AccessoriesIntraoperative Neuromonitoring (IONM) Electroecephalography (EEG) Electromyography (EMG)Nerve Conduction Study (NCS) Evoked Potential (EP)Axelgaard Stimulation ElectrodesTouch Proof Plugs Adaptors & Electrode Linkers
Cath SecureCATH-SECURE – OriginalCATH-SECURE – ExtendedCATH-SECURE PlusCATH-SECURE – Dual TabCATH-SECURE For KidsNG SECURE
PessariesPessary Ring Sizing KitsCerclage Pessary (Perforated)Cerclage Pessary (Non Perforated)Ring PessaryThick Ring PessaryCube Pessary (Perforated)Cube Pessary (Non-Perforated)Vaginal DilatorUrethra PessaryBowl PessaryUrethra Bowl PessarySieve Bowl PessaryHodge PessaryClub PessaryTandem Pessary (Perforated)Tandem Pessary (Non-Perforated)
Urodynamic ConsumablesUrodynamics CathetersPump Infusion SetsTransducer Pressure DomesFemale Voiding AdaptorDuckbill ValvesSetguards3-Way Taps
£1,395.00 exc. VAT
The HumBug Noise Eliminator is the ideal piece of biological research technology, aiding in the removal of mains or line noise (50/60Hz) in signal readings from biological sensors and high impedance devices, without the signal degradation associated with notch filters.
Need to remove noise from more than one channel?
Those of you wanting to remove mains interference from multiple channels can use more than one Humbug, but they are not inexpensive devices. Fortunately, Digitimer have developed our new D400 Two and Four Channel Main Noise Eliminators for cost-effective mains noise removal in multi-channel recording applications. The D400-2 and D400-4 remove mains noise from signals using a similar approach to the Humbug, but do so independently for several channels, even making it possible to share a single D400 across multiple rigs.
The Traditional Approach
A New Solution – 50 or 60Hz Noise Removal without Filtering
Publications which cite use of the Humbug can be found on Google Scholar.
The HumBug is NOT a mains noise filter, making this the main reason it is a special, perhaps unique product. Filtering often results in phase shifts, frequency loss, amplitude errors, DC shifts, time delays or digital distortion. The HumBug will eliminate 50/60Hz electrical interference (and harmonics) without altering your signal of interest.
A noisy waveform is the summation of two separate components. One component is the biological signal of interest and the other is noise of various forms. These forms of noise include random fluctuations generated by the sensor, amplifier and background biological noise in the vicinity of an experimental setup. The HumBug is designed to cancel electrical interference. It constructs a replica of noise embedded within the input signal and subtracts this replica from the noisy signal as it passes through the device. The end result is an output which contains only the signal of interest – mains noise removal without filtering.
The HumBug adapts rapidly when the input signal is dominated by noise and proceeds more slowly when low amplitude noise is embedded within continuous physiological activity. Adaptation is also slower for harmonics with frequencies greater than 1kHz.
Yes, but please remember that the HumBug is a single channel device, therefore you will need one HumBug for each channel of interest.
First you should check that the HumBug is configured to remove noise of the appropriate frequency. You can check this as it should be indicated next to the power connection socket on the back of the unit. It may be possible that the noise you are experiencing is not picked up from the mains supply. If the peak to peak noise is too large for the HumBug to completely cancel. If so, the indicator will continue to flash red. The HumBug cannot eliminate noise greater that 1V peak to peak. Likewise, low amplitude noise (<15mv) can be difficult to remove, so it is suggested that you increase the gain of your signal to ensure that the amplitude is greater than 15mV peak to peak.
It is possible, but not as easy as it seems to be at first glance. The problem is, that the EPC 10 has no Current In AD channel. The current signal from the amplifier board is hard wired with the AD converter inside the housing of the amplifier. Therefore it is not possible to connect the output of Filter 1 or Filter 2 via the HumBug and a Current In AD.
One would have to use one of the free AD inputs of the EPC 10 for reading the current signal. This has some drawbacks: the “real” current AD is automatically scaled by the “gain” that is set in the amplifier dialog. This is not the case for other AD inputs. As a consequence, you would have to scale the AD input manually (in the configuration dialog). This is no problem as long the gain is kept at the same value during an experiment. However, if the gain is changed frequently, then it might become annoying to change the AD scaling in the configuration many times.
Please leave this field empty.